Well-life TENS Machine
|Channel:||Dual, isolated between channels|
|Modes of Operations:||Burst, Continuous, Modulation|
|Pulse Intensity:||Adjustable 0-80MA peak into 500 ohm load each channel, constant current|
|Pulse Rate:||2Hz-150HZ (adjustable) 30uS-260uS (adjustable)|
|Pulse Width: Timer:||
|Timer:||Continuous, 30 min., 60 min.|
|Burst Mode:||Burst consists of 2 burst per sec at 100 Hz|
|Wave Form:||Asymmetrical Bi-Phasic square pulse|
|Voltage:||0-100 Volt (open current)|
|Power Source:||9 volts battery (Type 9V 6F22)|
Security policy (edit with Customer reassurance module)
Delivery policy (edit with Customer reassurance module)
Return policy (edit with Customer reassurance module)
TENS devices are clinically proven useful in pain management for many patients . By reading this manual and carefully follow the treatment instructions given to you by your physician / clinician , you will attain the maximum benefits from your TENS device .
Your TENS device may include the following components or accessories :
TENS unit Carry case Lead wires Electrodes Battery ( Type 9V 6F22 ) Operation manual
Mode Functions Burst ( B ) releases individual bursts twice per second , pulse width is adjustable and the pulse rate is set at 100Hz per second .
continuous ( C ) stimulation is delivered continuously at the settings determined by intensity , rate , and width knobs .
Modulation ( M ) pulse width decreases from its setting by 60 % and maintain the decreased width for 2 seconds before returning to the original width setting , which is maintained for 3.5 seconds . The cycle is then repeated . The intensity and pulse rate are adjustable .
NOTE : Use caution when you plug and unplug the wires . Pulling on the lead wire instead of its insulated connector may cause wire breakage .
CAUTION : Never insert the plug of the lead wire into an AC power supply socket
No customer reviews for the moment.