Meddina Well-life Tens Machine
|Channel:||Dual, isolated between channels|
|Modes of Operations:||Burst, Continuous, Modulation|
|Pulse Intensity:||Adjustable 0-80MA peak into 500 ohm load each channel, constant current|
|Pulse Rate:||2Hz-150HZ (adjustable) 30uS-260uS (adjustable)|
|Pulse Width: Timer:||
|Timer:||Continuous, 30 min., 60 min.|
|Burst Mode:||Burst consists of 2 burst per sec at 100 Hz|
|Wave Form:||Asymmetrical Bi-Phasic square pulse|
|Voltage:||0-100 Volt (open current)|
|Power Source:||9 volts battery (Type 9V 6F22)|
Security policy (Refer Term & Condition)
Delivery policy (Refer Term & Conditions)
Return policy (Refer Term & Conditions)
TENS devices are clinically proven useful in pain management for many patients . By reading this manual and carefully follow the treatment instructions given to you by your physician / clinician , you will attain the maximum benefits from your TENS device .
Your TENS device may include the following components or accessories :
TENS unit Carry case Lead wires Electrodes Battery ( Type 9V 6F22 ) Operation manual
Mode Functions Burst ( B ) releases individual bursts twice per second , pulse width is adjustable and the pulse rate is set at 100Hz per second .
continuous ( C ) stimulation is delivered continuously at the settings determined by intensity , rate , and width knobs .
Modulation ( M ) pulse width decreases from its setting by 60 % and maintain the decreased width for 2 seconds before returning to the original width setting , which is maintained for 3.5 seconds . The cycle is then repeated . The intensity and pulse rate are adjustable .
NOTE : Use caution when you plug and unplug the wires . Pulling on the lead wire instead of its insulated connector may cause wire breakage .
CAUTION : Never insert the plug of the lead wire into an AC power supply socket
No customer reviews for the moment.